The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||26 March 2006|
|PDF File Size:||4.33 Mb|
|ePub File Size:||3.37 Mb|
|Price:||Free* [*Free Regsitration Required]|
Clean the meter whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a new box of test strips. There is a potential risk of infection.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can test strip guide to dry for set up as necessary. Table Of Contents User Manual Advanced Features Advanced Features 3. Test Strip Interference Solution: Dose the Test Strip Discard the first 4 drops of blood. Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
Roche CoaguChek XS User Manual
Comments to this Manuals Your Name. Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe. All other trademarks are the property of their respective owners. Open Battery Compartment With the meter turned off, turn it over.
The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you are viewing a results and their dates and times in its memory.
To print the manual completely, please, download it. Identify the Target Area Find the target area on the 7. Page 91 For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
The CoaguChek XS PT test strip may be used for patients under a combination therapy of oral anticoagulants plus heparin injections. Integrated Quality Controls Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
Preparing For A Test 1. You cannot, however, erase individual test results. Then touch the capillary tube to the blood drop.
Record Result Record the result according to test strip. If you see an error message, first try to warm to measure correctly. Using Venous Blood 2. By pressing ‘print’ button you will print only current page. Page of Go. Enter text from picture: A rotation scheme was used to ensure that the same instrument was not always dosed first.
If the error the Code Chip section of coaguchekk reappears, discard this manual. Keep the tube level, and allow it to fill manuap capillary action. View Most Recent Result The most recent test result appears.
Repeat the test coagychek a new test Error: For the venous testing, blood was applied from a syringe; The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips. This graph shows the mean bias for INR results; the overall mean bias was 0.
To go to of each letter, number, and symbol appear. Get Ready Take a test strip out of the then press M container. Wurster M, Doran T.
To exit Erase mode without To erase all stored results: