The KODAK Intraoral X-ray system is equipped with a thermal safety system that prevents the generator from overheating in case of intensive use. Product, Model Kodak Intraoral X-Ray System, Catalog Numbers , , , XR — Carestream Health, Inc. Made in France Kodak Intraoral X-Ray System Performance. Operational ease. Safety. The Kodak intraoral x-ray system is designed for performance and safety.
|Published (Last):||3 April 2010|
|PDF File Size:||8.13 Mb|
|ePub File Size:||5.53 Mb|
|Price:||Free* [*Free Regsitration Required]|
Designed intraorall dental professionals seeking ease of use and reliability, it provides high quality results for accurate and safe diagnoses. Once the old solder removed, the wires soldered back with new solder do not overheat place the high voltage contact spring and insert the high voltage multiplier block inside the plastic cup containing the feedback resistor.
Please Mr Faustino can you send me the service manual and im interessed to inyour spare parts can you mail me suissedental x-rwy. Changes in ABO certification: December 21, ADA taps Dr.
Open the catalog to page 1. Unless Trophy designed it with certain service lifetime of each unit. Hello, Concerning the HV feddback resistor, I have 3 ceramic high voltage feedback resistors in case of emergency assistance, also have the high voltage block with the xray tube and multiplier, please contact for assistance details.
The voltage devider intraora be immersed in oil completely, no air bubbles present in oil! Other information will follow by mail. Hello,was there any sparks inside the tubehead untill now? In addition, the system emits koxak radiation to maximize the amount of useful x-rays and significantly reduce the levels of radiation. December 13, New law seeks to improve access to oral health care for underserved patients.
Advertise Increase Your Brand Awareness. Content you may be interested in: During the Thomas P.
December 3, Live Webinar: Live Webinar 19 Mar Send me a mail and I will send you a link to view my technical files or give you more information. It includes an intuitive kldak wheel for easy tooth and program selection, and automatically calculates the optimum exposure time and dose settings.
May 11, Business The best solution for IPR just got better As a part of its mission to always provide dentists with only the best solutions for improved patient care, ContacEZ continues to refine, improve and expand its product lines to make them safer and more effective than effort.
My presentations Profile Feedback Log out. In addition, the timer is pre-programmed to automatically calculate settings—so operating the unit has never been so simple. We think you have d-ray this presentation.
Used & Refurbished Dental Equipment
Chad Gehani, an endodontist, as its new leader. About project SlidePlayer Terms of Service.
Thanks, I’ll try to get the panel from Related Searches Flat panel detector Dental imaging software Medical radiology software Cephalometric analysis software Dental software Simulation software Analysis software.
I would recommend you not dismantle the ceramic resistor. I drill a 1 mm hole near anode’s intraorxl in the plastic intraorral of the tube to be sure that all air bubbles are eliminated.
Thanks, I will forward your email for those who need help on tube repairs and software. Live Webinar Stay out of Jail: As part of its growth plans and in order to better serve its customers, ASI has announced that it is adding local sales representatives to cover designated territories throughout the U.
Published by Dayna Stone Modified over 3 years ago. As a part of its mission to always provide dentists with only the best solutions for improved patient care, ContacEZ continues to refine, improve and expand its infraoral lines to make them safer and more effective than effort.
It shows no numbers just KV Error.
Kodak Intraoral X-Ray System – IMAGELEVEL – PDF Catalogs | Technical Documentation
Faustinio I am also having KV error and the system beeping. Equipped with a focal spot x-rayy 0. The best regsrds, Alex. The soldering on the resistor gets loose and this is the reason why so many of you get error Just check if it’s ok 13 Kohm and MOhn and solder unplagged wire to it.
Air is a reason of fail in repair. Sign In or Register.